Providing a secure airway management during general anesthesia could be problematic in some medical cases, especially when there is a risk of regurgitation and aspiration of the gastric content due to increased intragastric pressure. The current study aimed to test the applicability of two types of LMA in several animal species and to compare its effectiveness to the endotracheal intubation method in securing sealed airway respiration as an alternative to using endotracheal tubes. The study was conducted in dogs (n=33), cats (n=9), swine (n=9), rabbits (n=5), sheep (n=7) and roe deer (n=1). One or both types of laryngeal masks were used for each animal species: LMA Classic™-cLMA and LMA ProSeal™-PLMA. The assessment of each laryngeal mask was performed by determining the insertion technique, the possibilities of first-attempt insertion and malposition, the compliance with various animal species, ventilation time, cuff pressure, and sealing capacity. The highest LMA size compatility in dogs (23,87±14,30 kg) was size-3 in six and size-4 in forteen subjects; In swine (43,22±12,32 kg), size-4; In rabbits (3,84±0,36 kg) size-1; and in sheep (48,29±4,65 kg) size-3 and size-4. Ventilation time was highest in swine and roe deer (121,11±42,85 min and 300,00 min, respectively) and lowest in cat (28,33±16,96 min). First-attempt LMA insertion success was lowest in rabbits (60%), and highest in sheep and roe deer (100%). Malposition was with highest rate in rabbits (40%) and lowest in cat, sheep and roe deer (0%). Gastric reflux was most frequently observed in sheep (71,4%) and roe deer (100%). The usage of LMA in the veterinary anesthetic practice significantly improves airway management in animals during general anesthesia. The inflated LMA cuff does not prevent its disposition. Therefore, both the drain and respiratory tubes must be fixed. The usage of LMA in rabbits was associated with higher incidence of malposition and other complications. Our findings suggest that LMA designed for humans can be used for airway management in veterinary medicine.
If the inline PDF is not rendering correctly, you can download the PDF file here.
1. Sager, J. (2018). Supraglottic airway devices and tracheal tubes and stylets. In: K.G. Cooley, R.A. Johnson, Veterinary Anesthetic and Monitoring Equipment (pp. 177-178). Hoboken: Wiley Blackwell. https://doi.org/10.1002/9781119277187.ch14 PMid:29854387 PMCid:PMC5956637
2. McNarry, A.F., Patel, A. (2017). The evolution of airway management - new concepts and conflicts with traditional practice. British Journal of Anaesthesia 119(1): 154-166. https://doi.org/10.1093/bja/aex385 PMid:29161401
3. Fujita, M., Orima, H., Simazu, M., Motoyoshi, S., Katayama, M., Miyasaka, K. (1991). Use of laryngeal mask airway in small animals. J Vet Med Sci. 53(6): 1081-1082. https://doi.org/10.1292/jvms.53.1081 PMid:1790220
4. Brimacombe, J. (1995). The advantages of the LMA over the tracheal tube or facemask: a meta-analysis. Can J Anaesth. 42(11): 1017-1023. https://doi.org/10.1007/BF03011075 PMid:8590490
6. Kazakos, G.M., Anagnostou, T., Savvas, I., Raptopoulos, D., Psalla, D., Kazakou, I.M. (2007). Use of the laryngeal mask airway in rabbits: placement and efficacy. Lab Anim. 36(4): 29-34. https://doi.org/10.1038/laban0407-29 PMid:17380146
7. Reed, F., Iff I. (2012). Use of a laryngeal mask airway in a brachycephalic dog with masticatory myositis and trismus. Can Vet J. 53(3): 287-290.
8. Brain, A.I.J., Verghese, C., Strube, P.J. (2000). The LMA ‘ProSeal’-a laryngeal mask with an oesophageal vent. Br J Anaesth. 84(5): 650-654. https://doi.org/10.1093/bja/84.5.650 PMid:10844848
9. Brimacombe, J.R. (2005). Laryngeal mask anesthesia - principles and practice (p. 700). Edinburgh: Saunders.
13. Fulkerson, P.J., Gustafson, S.B. (2007). Use of laryngeal mask airway compared to endotracheal tube with positive-pressure ventilation in anesthetized swine. Vet Anaesth Analg. 34(4): 284-288. https://doi.org/10.1111/j.1467-2995.2006.00329.x PMid:17490468
14. Rokamp, K.Z., Secher, N.H., Møller, A.M., Nielsen, H.B. (2010). Tracheal tube and laryngeal mask cuff pressure during anaesthesia - mandatory monitoring is in need. BMC Anesthesiol. 10, 20. https://doi.org/10.1186/1471-2253-10-20 PMid:21129183 PMCid:PMC3016349