In the paper the concept of synthetic aperture used for high resolution/high frame rate ultrasonic imaging is reviewed. The synthetic aperture technique allows building extended “virtual” apertures, synthesized from smaller real aperture resulting in improved lateral resolution along full penetration depth without sacrificing the frame rate.
Especially, four methods, synthetic aperture focusing (SAF), multi-element synthetic aperture focusing (M-SAF), synthetic receive aperture (SRA) and synthetic transmit aperture (STA) are addressed. The effective aperture function, describing two-way, far field radiation is a useful tool in beam pattern analysis. Some basic notations, which are used to calculate the effective aperture are introduced in Appendix.
 KARAMAN M., LI PAI-CHI, O'DONNELL M. (1995),Fellow, IEEE Synthetic Aperture Imaging for Small Scale Systems, IEEE Transactions On Ultrasonics, Fer-roelectrics, and Frequency Control, 42, 3, 429-442.
 KARAMAN M., BILGE H., O'DONNELL M. (1998),Adaptive multi-element synthetic aperture imaging with motion and phase aberration correction, IEEE Trans. Ultrason. Ferroelec. Freq. Contr., 45, 4, 1077-1086.
 KARAMAN M., O'DONNELL M. (1998), Subaperture processing for ultrasonic imaging, IEEE Trans. Ultra son. Ferroelec. Freq. Contr., 45, 126-135.
 KARWAT P., KLIMONDA Z., SEKLEWSKI M., LEWANDOWSKI M., NOWICKI A. (2010), Data Reduction Method for Synthetic Transmit Aperture Algorithm, Archives of Acoustics, 35, 4, 635-642.
 KILMER M.E., MORAVITZ MARTIN C.D. (2004), Decomposing a tensor, SIAM News, 37, 9.
 KIM D., PARK J., CHO J., SONG T.K., YOO Y. (2012), New synthetic aperture imaging technique with dynamic apodization window, Proc. SPIE, Medical Imaging.
 LOCKWOOD G.R., FOSTER F.S. (1995), Design of sparse array imaging systems, Proc. Ultrasonics Symposium, 1237-1243.
 MOSEY S.A., CHARLTON P.C., WELL I.A. (2013), Modified Synthetic Aperture Focusing Technique Utilising the Spatial Impulse Response of the Ultrasound, NDT.net e-Journal.
 NIKOLOV M., BEHAR V. (2005a), Analysis and Optimization of Medical Ultrasound Imaging Using the Effective Aperture Approach, Cybernetics and Information Technologies, 5, 2, Sofia, Bulgarian Academy of Sciences, 53-68.
 NIKOLOV M., BEHAR V. (2005b), Analysis and Optimization Of Synthetic Aperture Ultrasound Imaging Using The Effective Aperture Approach, International Journal, Information Theories and Applications, 12, 3, 257-264.
 NOCK L.P., TRAHEY G.E. (1992), Synthetic receive aperture imaging with phase correction formotion and for tissue inhomogeneities. Part I: Basic principles, IEEE Trans. Ultrason. Ferroelec. Freq. Contr., 39, 4, 489-495.
 O'DONNELL M., THOMAS L.J. (1992), Efficient synthetic aperture imaging from a circular aperture with possible application to catheter-based imaging, IEEE Trans. Ultrason. Ferroelec. Freq. Contr., 39, 366-380.
 TRAHEY G.E., NOCK L.P. (1992), Synthetic receive aperture imaging with phase correction for motion and for tissue inhomogeneities. Part II: Effects of and correction for motion, IEEE Trans. Ultrason. Ferroelec. Freq. Contr., 39, 4, 496-501.
 TROTS I., NOWICKI A., LEWANDOWSKI M. (2008), Laboratory setup for synthetic aperture ultrasound imaging, Archives of Acoustics, 33, 4, 573-580.