The paper presents the results of finite element analysis and experimental testing under simulated physiological loading conditions on issues shaping the functional properties of individual all-ceramic abutments manufactured by CAD/CAM technology. The conducted research have cognitive significance showing the all-ceramic abutment behavior, as a key element of the implantological system, under the action of cyclic load. The aim of this study was evaluation the fatigue behavior of yttria-stabilized zirconia abutment submitted to cyclic stresses, conducted in accordance with EN ISO 14801 applies to dynamic fatigue tests of endosseous dental implants.
 C. Larsson, Swed Dent. J. Suppl 213, 9 (2011).
 W.N. Garine, P.D. Funkenbusch, C. Ercoli, J. Wodenscheck, W.C. Murphy, Int J Oral Maxillofac Implants 22, 928 (2007).
 Ch. Woong-Rae, H. Yoon-Hyuk, P. Chan-Jin, Ch. Lee-Ra, J Adv Prosthodont. 7(4) 288 (2015).
 A. Khraisat, O. Abu-Hammad, N. Dar-Odeh, A.M. Al-Kayed, Clin Implant Dent Relat Res. 6, 157 (2004).
 EN ISO 14801. Dentistry – Implants – Dynamic fatigue test for endosseous dental implants (2007).
 ISO 5832-2. Implants for surgery – Metallic materials. Unalloyed titanium (1999).
 ASTM F 136-08. Standard specification for wrought Titanium-6Aluminium-4Vanadium ELI (Extra Low Interstitial) alloy for surgical implant applications (UNS R56401) (2008).
 ISO 13356. Implants for surgery – Ceramic materials based on yttria-stabilized tetragonal zirconia (Y-TZP) (2008).
 J. Marciniak, J. Szewczenko, W. Walke, M. Basiaga, M. Kiel, I. Mańka, Biomechanical analysis of lumbar spine stabilization by means of transpedicular stabilizer in E. Piętka, J. Kawa (Eds.), Information Technologies in Biomedicine, Springer (2008).
 G.J. Lavigne, S. Khoury, S. Abe, T. Yamaguchi, K. Paphael, J Oral Rehabil. 7, 476 (2008).
 A. Ziębowicz, A. Kajzer, W. Kajzer, J. Marciniak, Metatarsal Osteotomy Using Double-Threaded Screws - Biomechanical Analysis, in E. Piętka, J. Kawa (Eds.), Information Technologies in Biomedicine, Springer (2010).