Evgeni Eltzov, Abri Lavena De Cesarea, ‘Yuen Kei Adarina Low and Robert S. Marks
; 238: p. 71-75.
94. Wan H, Yin H, Lin L, Zeng X,Mason AJ, Miniaturized planar room temperature ionic liquid electrochemical gas sensor for rapid multiple gas pollutants monitoring. Sensors and Actuators B: Chemical, 2018; 255: p. 638-646.
95. Kuberský P, Syrový T, Hamáček A, Nešpůrek S,Syrová L, Towards a fully printed electrochemical NO2 sensor on a flexiblesubstrate using ionic liquid based polymer electrolyte. Sensors and Actuators B: Chemical, 2015; 209: p. 1084-1090.
96. Rao Z, Liu L, Xie J,Zeng Y, Development of a benzene vapour sensor
Harsha Kasi, Robert Meissner, Alexandre Babalian, Harald van Lintel, Arnaud Bertsch and Philippe Renaud
resistivity is linearly dependent on the log PRF for both the species. For embryonic chicks, owing to similarity in the data across various trials, it can be observed that there is a unique PRF for each resistivity. On the contrary, the three rat trials suggest multiple PRFs for each resistivity.
PRF vs. resistivity plots for the five embryonic chick trials. In general, a good reproducibility of resistivities at a particular PRF in the trials was observed.
To the best of our knowledge, planar, bipolar microelectrodes on a flexible